|HOME --||SEQUENCING --||LIBRARY PREP --||HIGH-THROUGHPUT --||COMPUTING --||OTHER TECHNOLOGY|
The Covaris E220 evolution sonicator is a multitube system developed for high performance sample preparation. The E220evolution is capable of processing a wide range of sample types and volumes and may be programed to process from 1 to 96 samples in a single batch. The E220 is most appropriate <5ug clean DNA in 130ul (microTUBE-130) or 20-120 mg tissue mass in 1.0 ml (milliTUBE-1). Using adaptive focused acoustic energy, the E220 utilizing a spherical acoustic transducer, and delivers controlled energy precisely and accurately to sample volumes from 25 μl to 10 ml. The main applications are shearing of DNA, RNA, and chromatin but it can also perform tissue homogenization for cell lysis and drug delivery and formulation.
|Instrument||Covaris E220 evolution|
|TYPE||WALKUP - MIT only|
|UNIT||Per hour, Per Tube|
|FORMAT||Accepts covaris tubes and plates. |
Single point sonication only.
|CONSUMABLES||Many Covaris tubes are available to sign out in the BioMicro Center.|
|NEW USERS||New users should request training by emailing email@example.com|
|DONATED BY||Dept of Biology. |
Upgrade by Prof Manolis Kellis.
|Tube Type||Input/volume||batch size||Available at BMC?|
|microTUBE plate, strip, or crimp cap||5ug/130ul||1-96||Bring your own|
|microTUBE-500||truXTRAC® FFPE RNA||1-24||Yes|
|Currently unavailable platforms|
|If you want to use one of these - let us know|
(BE CAREFULL WITH the ACOUSTIC ASSEMBLY LENS, REPORT IMMEDIATELY if SHARPS (like broken glass) GOT ONTO IT)
Two types of maintenance after Covaris use:
Case#1. Covaris was connected to working water-filtering unit;
Case#2. Covaris was not connected to water-filtering unit (hosing is taken off, water-filtering unit is switched "OFF".
Maintenance Case#1(usually, water-filtering is set up for multiple Covaris usage, degas-ready for all subsequent runs, after the 1st)
After your run is finished - a pop-up alert will appear on the computer screen.
(1)-Click "OK", then
(2) click "Load" button to move the holder to "Load" position,
(3) take the rack containing your samples out of COVARIS chamber,
(4) close the chamber,
(5) switch off the chiller unit (underneath the Covaris)
(6) click "Service" button to raise the Sample Holder and Acoustic Assembly above the water bath. Degas pump will stop.
(7) Quit COVARIS software, check that the tubing connectors of COVARIS (front panel) to water-filtering unit (front panel) are not leaking any water,
(8) clean the area around the machine, wipe off any water spills.
(9) Optional: turn the Covaris "OFF" (switch is on the right side of the machine), - lights inside the machine will go off,- after you press the switch.
(10)Optional: Switch the water-filtering unit "ON"(right side of the Covaris Instrument)
If you used Covaris without connection to Water-Filtering Unit - you must empty water bath after use.
(A) Follow the instructions for Case#1 to the step (6)("Sevice Position"). It raises the acoustic assembly above water bath and enables it;' removal
(B) Gently and carefully pull out (towards you) the plexiglass bath of Covaris (currently filled with water) - and empty it into the sink, then - reinstall the bath.
(C) re-install the empty bath, click "load position": the acoustic assembly will go down
(D) temporarily, for few seconds, switch the degas pump "ON": it will purge some extra water out into the bath - and stop. If not stopped in 10s - click "degas OFF"
(E) Quit the Covaris software and turn off the Covaris machine (upper-right side button. Make sure the Chiller Unit and Filtering unit are OFF. Cleanup any spills
The Covaris service is now complete.