Microfluidic Sensing- Microfluidic Biosensors- Xiao Fan
Biosensor is a kind of analytical device which combines biological sensitive recognition element (such as nucleic acids, enzymes, antibodies) with a physicochemical detector for analytes analysis in a sample.1 The integration of microfluidic device and biosensor technologies provides new opportunities for biosensing applications, such as portability, real-time detection, increased sensitivity and selectivity.2 A lot of microfluidic devices have been developed for biosensor applications. DNA-based microfluidic biosensors, enzymes-based microfluidic biosensors, microfluidic immunosensors are the major three kinds of microfluidic biosensors.
Video: Microfluidic Biosensors, New Frontier of Diagnosis
DNA-based microfluidic biosensors
DNA-based microfluidic biosensors usually immobilize complementary sequences of target nucleic acid in the microfluidic device. Then, samples with target nucleic acid are introduced in the microfluidic channel and the detection process can be achieved in microfluidic device. The recognition process relies on the base pair of hybridization probes and target nucleic acid in the sample. Base pairing process usually generates physicochemical signals, which can be used to analyze the target DNA in the sample.
Generally, detection of DNA involves the amplification of the sample DNA via polymerase chain reaction (PCR) and the subsequent detection of DNA fragments based on their electrophoretic mobility on a gel substrate.3 By developing microfluidic device, polymerase chain reaction (PCR) and the subsequent detection steps can be integrated into some microfluidic platforms, which reduces the risk of sample loss and contamination in general process.
Figure 1 shows an overview of the Integrated Microfluidic Electrochemical DNA (IMED) sensor, which integrates three biochemical functionalities for DNA sensing—symmetric PCR amplification, enzymatic single-stranded DNA generation and sequence-specific electrochemical detection. The microfluidic device is fabricated by using a PDMS layer and two glass wafers (as shown on the right side in Figure. 1A). In the sensing process, DNA and PCR reagents are introduced into the PCR chamber to rapidly amplify DNA. During PCR, the reverse primers generate phosphorylated strands, which are subsequently selectively digested by lambda exonuclease to efficiently produce ssDNA targets for detection. Then, MgCl2 is added for optimal hybridization and sequence-specific electrochemical detection is performed with electrochemical DNA (E-DNA) probes, where target hybridization causes a change in the redox current of the probe, which can be detected via alternating current voltammetry (ACV). The detection limit is less than 10 aM when using this microfluidic device to detect genomic DNA from Salmonella enterica serovar Typhimurium LT2.4
Enzyme-based microfluidic biosensors
Enzymes are proteins that catalyze chemical reactions. It is a kind of popular bioreceptors used in microfluidic biosensors because of their specific binding capabilities and catalytic activity. In an enzyme-based microfluidic biosensor, enzymes are typically immobilized onto a suitable transducer in microfluidic device that produces a specific signal (optical signal, electrochemical, colorimetric etc.) upon reaction with a given analyte in microfluidic channel.5 These signals could be used to obtain some important parameter of human body, such as cholesterol and blood sugar which have been found to connect with some diseases of human.
An example of an enzyme-based microfluidic biosensor is shown in Figure 2. A droplet-based microfluidic biosensor is fabricated based on poly (dimethylsiloxane) (PDMS) using soft lithograph. Glucose oxidase (GOx) is as the analyte, which catalyzes the oxidation of β-D-glucose to the product H2O2. In the testing process, the sample stream is combined with enzyme GOx and substrate glucose and then segmented into a serial of droplets by two intersecting streams of mineral oil. The enzyme activity of GOx can be evaluated by measuring the concentration of reaction product hydrogen peroxide. Since hydrogen peroxide is electroactive, the enzyme activity of GOx is electrochemically detected based on measuring the electrochemical current on a Pt-black microelectrode responding to various glucose concentrations. This device provides a highly sensitive and low-cost glucose sensor with linear response up to 43.5 mM. 6
Microfluidic immunosensors are based on the immunochemical reactions of antigen and antibody. Usually, antibody is spatially in contact with a transducer surface as recognition element to identify the antigen in the sample. Enzyme linked immunosorbent assay (ELISA) is the most widely used method for detection. Additionally, some microfluidic immunosensors can detect immunocomplex formation by changes in current, resistance, refractive index, etc.
Figure. 3 shows a microfluidic immunosensor using microfluidic enzyme linked immunosorbent assay (ELISA) platform for detection of antigen. The device is fabricated by three-layer PDMS/glass and contains four parallel assay units. In the sensing process, the capture antibody (Ab1) was first flowed through the assay chamber where the surface was silanized with APTES to enhance the immobilization of Ab1 through electrostatic interaction between the negatively charged antibodies and the positively charged amine group of 3-aminopropyl triethoxysilane (APTES). The antigen can be captured, recognized by the biotinylated detection antibody (BAb2), and quantified using b-galactosidase (b-gal) as the reporter enzyme for chemifluorescence detection.7
Figure.4 illustrates another example of immunosensors using antibodies as the biological recognition element for the detection of antigen. The microfluidic chip was fabricated by using three layers of PDMS and a glass substrate. In the testing process, the microfluidic chip is coupled with the SPR phase imaging system. Anti-rabbit IgG solution is constantly pumped through the microfluidic channel for immobilization. Then the rabbit IgG sample was injected into the microchannels. Interaction between rabbit IgG with immobilized anti-rabbit IgG have been successfully measured by using employing SPR phase imaging. The relation between phase difference and the concentrations of rabbit IgG can be obtained. The detection limit could be 1×10-4 mg/ml (0.67 nM).8
1 Nguyen, H.H., Lee, S.H., Lee, U.J., Fermin, C.D. and Kim, M., 2019. Immobilized enzymes in biosensor applications. Materials, 12(1), p.121. https://doi.org/10.3390/ma12010121
2 Luka, G., Ahmadi, A., Najjaran, H., Alocilja, E., DeRosa, M., Wolthers, K., Malki, A., Aziz, H., Althani, A. and Hoorfar, M., 2015. Microfluidics integrated biosensors: A leading technology towards lab-on-a-chip and sensing applications. Sensors, 15(12), pp.30011-30031. https://doi.org/10.3390/s151229783
3 Choi, S., Goryll, M., Sin, L.Y.M., Wong, P.K. and Chae, J., 2011. Microfluidic-based biosensors toward point-of-care detection of nucleic acids and proteins. Microfluidics and Nanofluidics, 10(2), pp.231-247. https://doi.org/10.1007/s10404-010-0638-8
4 Ferguson, B.S., Buchsbaum, S.F., Swensen, J.S., Hsieh, K., Lou, X. and Soh, H.T., 2009. Integrated microfluidic electrochemical DNA sensor. Analytical chemistry, 81(15), pp.6503-6508. https://doi.org/10.1021/ac900923e
5 Kumar, S., Kumar, S., Ali, M.A., Anand, P., Agrawal, V.V., John, R., Maji, S. and Malhotra, B.D., 2013. Microfluidic‐integrated biosensors: Prospects for point‐of‐care diagnostics. Biotechnology journal, 8(11), pp.1267-1279. https://doi.org/10.1002/biot.201200386
6 Gu, S., Lu, Y., Ding, Y., Li, L., Song, H., Wang, J. and Wu, Q., 2014. A droplet-based microfluidic electrochemical sensor using platinum-black microelectrode and its application in high sensitive glucose sensing. Biosensors and Bioelectronics, 55, pp.106-112. https://doi.org/10.1016/j.bios.2013.12.002
7 Wang, T., Zhang, M., Dreher, D.D. and Zeng, Y., 2013. Ultrasensitive microfluidic solid-phase ELISA using an actuatable microwell-patterned PDMS chip. Lab on a Chip, 13(21), pp.4190-4197.https://doi.org/10.1039/C3LC50783A
8 Lee, K.H., Su, Y.D., Chen, S.J., Tseng, F.G. and Lee, G.B., 2007. Microfluidic systems integrated with two-dimensional surface plasmon resonance phase imaging systems for microarray immunoassay. Biosensors and Bioelectronics, 23(4), pp.466-472. https://doi.org/10.1016/j.bios.2007.05.007