Difference between revisions of "User:Pranav Rathi/Notebook/OT/2010/04/01"
(→High power beam profiling)
|Line 15:||Line 15:|
Revision as of 13:05, 15 November 2012
|Project name||<html><img src="/images/9/94/Report.png" border="0" /></html> Main project page|
<html><img src="/images/c/c3/Resultset_previous.png" border="0" /></html>Previous entry<html> </html>Next entry<html><img src="/images/5/5c/Resultset_next.png" border="0" /></html>
High power beam profiling
Today I discovered a new way to profile a beam at high power. I used a regular 1064nm mirror. The mirror gives a 99.99% reflection at 45 degrees. So I used the mirror in between CCD camera and the AOM at some angle less than 45. In this way transmission through the mirror can be controlled. its highest at 0 and lowest at 45. In between desired transmission can be achieved. I received a transmission of 99mW at 4.10W at the angle of roughly 15 degrees. I used the CCD with 3+3+2 OD filters. This setup does not distort the beam profile too much and it’s safe for the OD filters. We got some good pictures through this process; the link is presented below. Steve Koch 18:46, 1 April 2010 (EDT): Great idea! Very cool that you can adjust the power like that. And great that you discovered a good fix for the laser issue! See Anthony's page for more suggestions. Also, "gernal" is spelled wrong. Did you mean "general" or "journal?"
Go to the link for detail view: