|Project name||<html><img src="/images/9/94/Report.png" border="0" /></html> Main project page|
<html><img src="/images/c/c3/Resultset_previous.png" border="0" /></html>Previous entry<html> </html>Next entry<html><img src="/images/5/5c/Resultset_next.png" border="0" /></html>
Tweezer is aligned
Our tweezer is aligned, but with an issue; as you can see there is more power to the left. I tried to fix this issue as much as i could. File:Tweezer 03-25-10.avi
I also made a new tool which will help us aligning the beam at the two end-apertures of the cage system. It's just a web-cam on a post, but web-cam captures the IR pretty well. I am planning to have two web-cams on the two ends so we can align the both ends simultaneously.
Power drift: I noticed that there is substantial power drift over the tweezer. I made this video at 2.78 W, with the reflection out of air, so I was not using oil or the glass slide. This video is around 3 minutes long and the change is obvious. I am still investigating the issue because I strongly believe that this is not due to power. File:Power drift.avi