Difference between revisions of "McClean:Equipment"
|Line 1:||Line 1:|
[[| Nikon Eclipse TI]]
Revision as of 16:12, 23 October 2012
Add here a list of equipment and maintenance logs for that equipment + reminders, etc.
Note on the Thermo -80C Freezer from Dipen:
Hi Megan, Jean
The technician from Thermo had a look at the freezer door today. The door was apparently getting locked up because the sensor that allows the equilibration of temperature after the door is closed, was frozen over. We should be able to take care of this problem in the future; there is a hole in the inside part of the door that needs to be periodically scraped out, particularly in the humid summer months.