User:Pranav Rathi/Notebook/OT/2010/05/12/Characterization of AOM module
The motive of this is to study AOM as power modulator for the tweezers. AOM (Gooch & Housego R23080-2-1.06-LTD; 138252) operates in CW and normal mode. We use 1st order diffracted beam. First CW mode operation is characterized, to know the usable power (of the diffracted beam) Vs ascending input laser power. Second normal mode operation is characterized: Computer controlled ascending input voltage for RF signal power Vs usable power of the diffracted beam, at several input laser powers. The laser temperature is kept constant at 62F.
Relatively simple setup ThorLabs power meter is place infront of the diffracted beam from AOM. An aperture is used infront of the power meter to keep off all the stray beams.
CW mode operation characterization:
The AOM is operating in CW mode. Data is recorded; laser power (.25W to 4W in .25W increments) Vs 1st order beam power. This 1st order beam power is the usable power for tweezer. The data is presented below:
The relationship is linear. So the AOM gives output power in 1st order diffracted beam, of average around 70% of the input power (in CW mode). This help us calculating the usable power of any input power without measuring; for example if I set the laser on .10W, then the usable power at the tweezer is 70% of it, which is .07W or 70mW.The maximum usable power for the tweezer is 2.66W. If we want to stay in the single mode operation regime, it is 1.4W (at room temperature). When the laser is cooled down to 60F, it is stretched to 1.9W (2.75W input).
Normal mode operation characterization
In normal mode, AOM is controlled through NI-DAQ by LabView program. Voltage input for RF signal is from 1 Volt to 5 Volts in any Volt increment. 1st order diffracted beam power is recorded for every ascending voltage increment of .1Volt for .5W, 1.5W, 2W and 2.5W input laser powers. Different laser powers are used to check: If the characteristic of AOM is input power dependent, which it should not be. RF-input voltage Vs 1st order diffracted beam power will help us in obtaining the best workable range of voltages over which power increases linearly. It will also help us in characterizing the relationship between the voltage and diffracted power. This is very important in data analysis. While data acquisition we can not measure the laser power in trap its is unpractical but if know the characteristic RF-input voltage Vs laser power in trap we can calculate the laser power in trap from the voltage. The data is presented below:
- Data Sheet 1 and Chart 1: By looking the chart it’s clear; diffracted beam power is not linearly related with voltage. It has strange characteristic, which is same for all laser powers: So there is no laser power dependence and any laser power can be chosen to define the AOM characteristic. Characteristic: It runs somewhat exponentially in the beginning from 1 to 1.4 volts, somewhat linear between 1.5 to 3.1 volts (the two straight vertical lines on the chart represents that) and second degree polynomial in the end from 3.2 to 5 volts. So the most appropriate workable range for linear power modulation (in relatively same size of steps) is from 1.5 to 3.1 Volts at any input laser power.
- Data Chart3: Shows how power increases in % over 1 to 5 Volts at 2W of laser power. By joining the head of each histogram the characteristic curve can be traced out. Between 1.5 and 3.1 Volt the power increases almost in same step size of avg 3.45% (.069W) /step in comparison to .8% (not linear; over range of 1 to 1.4 Volts) and .42% (not linear; over range of 3.1 to 5 Volt). 3.45% step size is more convenient for power modulation and power is also linear over the steps.
- Data Chart 2: Shows power step in %; the difference between the two consecutive power-increments over the difference in the two consecutive voltage increments (.1 Volt; steps). It gives us the better idea of over what range of voltage, the steps are over all large with somewhat of same histogram height difference. The region between 1.5 to 3.21 Volts contains that.