Back to Protocols
- If you desire 4oC during your spin cycle, you must allow the machine to cool to that point before it will allow you to press start.
- If you use too small of a volume in the ultracentrifuge tubes, they will collapse!
- It is hard to say how much is enough, but it seems to be about 1 mL. 1/2 mL definitely collapsed on Janet in 5/2013.
- We should be able to find smaller-volume tubes for our rotor. (link)
- You probably won't be able to get any sample out of a collapsed tube.
- When they collapse it is very tricky to get them out. Needle nose pliers are probably your best bet but it can still be very hard.
- If the ultracentrifuge won't start and won't even try to start, open the door, lift the rotor up, place it back down, let it cool back down to your desired temp, and hold start.
- To spin down the membranes and membrane-bound proteins in a crude extract, centrifuge for 45,000 RPM at 4oC for 1 hour. With our MLS-50 mushroom-shaped rotor, this is 163,000xg at raverage (108,000xg at rminimum).