Image:Detector Wiggles for Various AOD Powers.png

The units on this graph are a little confusing for legacy reasons. I think the X-axis is the AOD frequency plus some offset. Basically it relates to the optical tweezers position in the sample plane, full scale is like 10 or 20 microns left to right. At the bottom left you can see the scale in nanometers, all the traces are scaled equally, so at high AOD power (which is low DVE voltage, towards the top), the wiggles are really bad, close to 10 nm ?