Quantification of glucose

SET-UP

 * 1) Rinse out Buffer container and fill to line with YSI 2357 Buffer solution
 * 2) Hit [Menu} then {2-MeasParameter] then [2-CalMethod] then [1-One Station]
 * 3) Hit [Enter] twice to return to Measurement Parameter Set up Screen
 * 4) Hit [3-Black]
 * 5) Make sure Current BLACK probe chem is set to NONE.  If it isn't, hit [1-Next chemistry] until NONE appears and hit [ENTER] to save
 * 6) Hit [4-White] and set Current WHITE probe chem to Dextrose
 * 7) Hit [Enter] 4 times to return to Measurement Parameter Set up Screen
 * 8) Hit [Menu] to return to home screen

RUN SAMPLES

 * 1) Hit [Run] and wait for system to complete calibration
 * 2) When Ready screen appears, hit [Sample]
 * 3) Present sample to needle in Station S3 without touching needle to tube
 * 4) Hit [Sample] and wait for needle to retract
 * 5) Results will print out at the completion of run
 * 6) To continue testing samples, hit [Sample] when Ready screen appears


 * Note: After 5-7 runs, system will automatically calibrate.