Physics307L F09:People/Barron/Final final

Speed of Light from a Cardboard Tube
Alexander T. J. Barron

Experiment conducted with Justin Muehlmeyer

Junior Lab, Department of Physics & Astronomy, University of New Mexico

Department Website

December 14, 2008 

Abstract
I measure the speed of light in air utilizing a time-to-amplitude converter (TAC), a photo-multiplier tube (PMT), and an LED pulse-generator in a light-tight environment. By positioning the pulse-generator at varying distances from the PMT/TAC apparatus, one can obtain various data sets corresponding to the change in distance (Δx) between the two components. For each Δx, the TAC manifests a new amplitude, corresponding to change in time (Δt) readings. Plotting Δx vs. Δt yields the perfect environment for taking a least-squares linear fit of the data, of which the slope is the speed of light in air. In the process of finding the speed of light in air, I investigate the comparative effectiveness of varying rates of change in Δx. I also investigate a phenomenon known as "time walk," which introduces systematic error significant enough to render the experiment unreliable. I address and minimize this phenomenon to obtain experimental values statistically comparable to the accepted value of the speed of light in air.



Introduction
Even after the Michelson-Morley experiment in 1887 gave reasonable doubt as the existence of the aether, certain scientists argued into the early 20th century against the case for (what we now call) relativity, based on statistical uncertainty Heyl-Science-1911 Young-Freedman. Least-squares analysis specifically was targeted as covering up true values in its effort to smooth out errors, thereby burying results leading to aether-positive results. Through Einstein's theory of special relativity, and corroborating evidence Heyl-1926 Hodgson-1978, we now know almost irrevocably that the aether is not a factor in measurements of the speed of light, so good experiments involving least-squares analysis can be pursued with impunity.

All experiments measuring the speed of light prior to 1944 involved taking measurements of time taken for light to traverse a given path Dorsey-Transactions-1944. The most common method up to 1944 was to partition light "packets" with periods of zero luminosity, thereby creating specific boundary times with which to measure between Dorsey-Transactions-1944. Such methods are labeled "time of flight" experiments. Since then, more sophisticated experiments successfully corroborated and improved upon the accuracy of older values. Lasers were at the forefront of the new experiments, which used observations of frequency to more clearly define the speed of light Blaney-1974 KD-1972.

The contemporary accepted value of the speed of light is 299,792,458 m/s BIPM-2006, as is defined by the specified measure of a meter, combined with the standardized measurement of a second by the Bureau International des Poids et Mesures(BIPM). This value reduces to 2.997 E 08 m/s in air at STP.

Recently, experimentalists have performed experiments aiming to challenge the canonical speed of light, trying to bend the seemingly impenetrable causality barrier resulting from the velocity maximum currently understood. The two extremes of such experiments consist of either slowing the speed of light down to an infinitesimal fraction of its normal magnitude through various creative physical media, or speeding up the group velocity of light pulses to find a speed greater than the standard Wang-2000. Such experiments do not threaten the accepted theory of special relativity and causality, however, as no physical laws are broken via such means Ball-2000.

Even in view of all the advances in experimental methods to determine the speed of light, careful data-taking in well-designed time-of-flight experiments can yield results statistically very close to the standardized value. I accomplish this via a simple experimental setup, outlined below, and a statistical least-squares analysis.



Methods and Materials
In order to measure the time-of-flight (TOF) of light packets, we positioned a moveable LED-pulse generator (custom) inside a several meter-long cardboard tube. The generator filled the entire cross-sectional area of the tube. On the opposite end of the tube, we positioned a fixed PMT (Perfection Mica Company, #N-134), which also filled the cross-section, ensuring light-tight conditions. On the inner side of each device was a polarizer, used to maintain near-constant intensity of light received by the PMT. Constant intensity was needed to minimize the effects of "time-walk," addressed under Sources of Error below. The generator and PMT were both connected to the TAC, which measured the difference in time between the generation of the pulse and the receipt of the pulse by the PMT. The PMT was also connected to a digital oscilloscope (TEKTRONIX TDS 1002) through a second anode connection, in order to maintain a reference voltage needed to counter the effect of time-walk. We read the voltage output of the TAC via the oscilloscope along with the reading from the PMT (Figure 1).

With this setup, we were able to take data over a range of varying distance parameters, denoted in the following four trials.

i) large and increasing individual Δx over large total Δx,

ii) small, constant individual Δx over small total Δx,

iii) large, constant individual Δx over large total Δx, and

iv) medium Δx with no time walk correction.

For each Δx, we moved the LED-pulse generator towards the PMT in intervals specified under Results below (Table 1).

Sources of Error
Time walk was the principle source of systematic error in this experiment. It occured due to the TAC's triggering off signals from the PMT via a fixed voltage threshold. If the signal from the PMT was small, the TAC would trigger later than it would for a larger signal. In order to combat this, we used rotatable polarizers in tandem to try and maintain constant intensity of light received by the PMT. The higher the intensity, the more photons interacted with the PMT and a stronger signal resulted. The same was true in reverse.

We read the voltage amplitude from the TAC visually using cursors on the digital oscilloscope, so our measurements were not very precise. We were aided by the averaging tool provided by the oscilloscope, which helped in reading data from the oscilloscope screen.

Our device for measuring distance was a standard meterstick, whose accuracy compared to the standardized meter was not documented.

Results
Measurements taken over a large individual and total Δx, as in trials 1 and 3, yielded the best results for the speed of light in air (Table 2). Unfortunately, the length of our experimental apparatus was limited, so measurements taken in consecutive movements of the meter stick limited the amount of data taken in this way. This, in turn, resulted in a large standard error. A differing method could be followed, allowing Δx measurements to be taken non-consecutively, which probably would have reduced the standard error via increased number of data. Measurements taken over small individual and total Δx as in trial 2 yielded a mean value inconsistent with the best two trials, with a standard confidence interval inconsistent with the accepted value (Figure 2). This effect is probably due to the significant amount of systematic error overwhelming the trend of light propagation observed over such a small changes in distance. Measurements taken over large total distances probably influenced the data so that the true trend could overcome systematic error unchecked in smaller-distance measurements.

The result for trial 4 illustrates quite clearly the importance of time walk correction in this experiment. The systematic error created from neglecting this phenomenon resulted in accuracy at least 4 σ away from the accepted value. Despite a fairly precise data set, which narrowed error below that of better measurements, the final value via trial 4 is unacceptable.



Conclusion
Upon comparison with the accepted value for the speed of light in air, this experiment seems fairly sound. Correction of time-walk was essential to garnering quality data, and was the chief source of systematic error. Other sources were not thoroughly investigated, but manifested themselves in sufficient quantity to make one data trial unacceptable. The experiment could be improved further with the use of a data acquisition card or another computer interface to measure voltage from the TAC. Use of experimental controller programs such as Labview would enable highly accurate monitoring of the reference voltage, which would reduce the effect of time-walk more effectively. Individual values of TAC voltage would could be correlated with the reference voltage automatically through the program interface, so that only high-quality data would be taken. Individual voltage measurements taken this way would provide a solid data set for a more accurate statistical analysis. A higher-quality device for measuring distance would realize more precise measurements, improving statistical analysis, as well as improving accuracy via calibration to the accepted unit value of the meter. 

Acknowledgements
Thanks to Miles Davis, Madonna, and Psapp, who really spiced up the data analysis. A big thank you to all the lab equipment- they're the real unsung heroes. Thanks finally to Justin, go team, Dr. Koch, go open science, and Aram, go physics!